SepaBean™ machine 2
|Model||SepaBean™ machine 2|
|Detector||DAD variable UV (200 - 400 nm)||DAD variable UV (200 - 400 nm) + Vis (400 - 800 nm)|
|Flow Range||1 - 300 mL/min|
|Maximum Pressure||500 psi (33.5 bar)|
|Pumping System||Highly accurate dual piston pumps|
|Gradients||Four solvents binary with 3rd solvent as modifier|
|Sample Loading Capacity||10 mg - 33 g|
|Column Sizes||4 g - 330 g, up to 3 kg with adapters|
|Gradient types||isocratic, linear, step|
|Flowcell optical path length||0.3 mm (default); 2.4 mm (optional)|
|Sample loading method||manual load|
|Fraction collection method||all, waste, threshold, slope, time|
|Fraction collector||Standard: tubes (13 mm, 15 mm,18 mm, 25 mm);
Optional: French square bottle (250 mL, 500 mL) or large collection bottle;
Customizable collection container
|Control device||wireless operation through mobile devices*|
Wireless Operation Through Mobile Devices
The flexible wireless control method is especially suitable for separation experiments that need to be protected from light or placed in an isolator.
Power Failure Recovery
The built-in power-off recovery function in the software minimizes the loss caused by accidental power failure.
Separation Method Recommendation
The software has a built-in separation method database that automatically recommends the most appropriate separation method based on the key information entered by the user, thereby improving work efficiency.
Tube racks with LCD display enable users to easily track the tubes containing collected fractions.
Local Network Data Sharing
Multiple instruments could form a local area network to facilitate internal data sharing and resource optimization in the laboratory.
21-CFR Part 11 Compliance
The control software complies with FDA requirements for system safety (21-CFR Part 11), making the instrument more suitable for pharmaceutical R&D companies and laboratories.
Smart purification system makes the purification easier
The smart flash chromatography system SepaBean™ machine 2 launched by Santai Technologies has the built-in feature of separation method recommendation. Even the beginners or non-professional chromatography operators could easily complete the purification task.
Smart purification with “Touch & GO” simplicity
SepaBean™ machine 2 is operated through mobile device, with iconized UI, it is simple enough for the beginner and non-professionals to complete routine separation, but also sophisticated enough for the professional or guru to complete or optimized a complex separation.
Built-In Method Database — Knowledge Retained
Researchers around the world spent numerous resources to develop methods of separating and purifying compound mixtures, whether it’s synthesized mixtures, or extracts from natural products, these valuable methods are usually stored in single location, isolated, disconnected, and become “information island” over the time. Unlike traditional flash instrument, SepaBean™ machine 2 employs database and distributed computing technology to retain and share these methods across secured organizational network:
● Patented SepaBean™ machine 2 has built-in relational database to store separation methods, researchers can query existing or update new separation method simply using compound name, structure or project code.
● SepaBean™ machine 2 is network ready, multiple instruments within an organization can form a private channel, so that separation methods can be shared across the entire organization, authorized researchers can access and run these methods directly without having to re-develop the methods.
● SepaBean™ machine 2 can discover and connect to peer instrument automatically, once multiple instruments are connected, data is automatically synced, researchers can access their methods in any connected instrument from any location.
AN014_The Improvement of the Resolution by Column Stacking and Its Application
AN015_Hydrophobic Phase Collapse, AQ Reversed Phase Chromatography Columns
AN016_The Application of SepaFlash™ Strong Cation Exchange Chromatography Columns in the Purification of Alkaline Compounds
AN018_The Application of C18AQ Columns in the Purification of Strong Polar Peptides
AN019_The Purification of Highly Polar Impurities in Antibiotics by C18AQ Columns
AN021_The Application of Column Stacking in the Purification of Organic Optoelectronic Materials
AN022_Higher Sample Loading Capacity, Better Performance – The Application of SepaFlash™ Ruby High Resolution Cartridges
AN032_The Purification of Diastereomers by SepaFlash™ C18 Reversed Phase Cartridge
AN-SS-002 Separation of CBDA and THCA from Cannabis and its relevance to biotechnological cannabinoid production
AN-SS-003 Facile Purification of large-scale steric selected bicyclic carbohydrates by SepaBean™ machine
AN-SS-004 Rapid Isolation Procedure for Δ9-tetrahydrocannabinolic acid A (THCA) From Cannabis sativa L. Using SepaBean™ Flash Chromatography Systems
AN-SS-005 Extraction Method Development for Cannabidiolic Acid from Cannabis sativa L. Using SepaBean™ Flash Chromatography Systems
Sepabean machine 2 Operation guide
SepaBean Device Setting – Tube Rack Calibration
SepaBean Maintenance — Nozzle Clean
SepaBean Maintenance — Air Purge
SepaBean Maintenance – Pump Calibration
SepaBean Machine Catalog EN